Will this radar device work with Dewe 43?
I'm looking at a ground speed radar device manufactured by Parker:
But will it work with our Dewe-43? How would we go about configuring Dewesoft to read the square wave output ( 58.9 Hz/mph )?
I've looked at the document that you've linked and I believe that the sensor should be compatible with the DEWE-43. You can use the counter's 12 V output to power the sensor and then feed its square output back into the counter module. You will then receive a frequency channel based on the input square. With that, you can use additional formulas to calculate the velocity or whichever parameter is supposed to be the final result of the sensor.
That being said, please understand that I did not thoroughly study the documentation so there's always the chance that I missed an important detail somewhere.