Setting up interface Load cell with DEWE-43A
Customer Support Engineer
On the picture below I have made a connection diagram. In the datasheet, it is stated only the maximum allowed voltage (20V) and no lower limit.
Dewe-43 can output max 12V and 350 mA. This power is shared between all the channels. If the sensor is too powerful for the Dewe-43 to supply it an additional power supply will be needed.
The EXC and SNS lines can also be used instead of the +12V, but they can supply only 10V and 70 mA per channel.
If you require any further information, feel free to contact us.