AC source for LVDT
Hey does anyone know if X3 can be used as an AC source to power the primary coil of an AC-AC LVDT? 3vrms and 3kHz. Any help is much appreciated!
For AC-AC LVDT sensor probes you would require one of ours DSI-LVDT adapters.
Adapter will provide AC Excitation for the LVDT sensor probes.
Please refer to technical reference manual (see attachment) for details.
Our DSI-LVDT adapter can provide only two preset frequencies:
Sensor Exc. voltage : 2,88Vrms (differential)
Sensor Exc. frequency : Selectable 4,02kHz / 9,60kHz typical; 2,5% error